A Homemade Viscometer I
Stirring a bowl of honey is much more difficult than stirring a bowl of
water. But why? The mass density of the honey is about the same as that
of water, so we aren’t moving more material. If we were to write out
The reason that honey is harder to stir is of course that the drag on our spoon depends on more than just the density of the fluid. The drag also depends on the viscosity of the fluid — loosely speaking, how thick it is — and the viscosity of honey is about 400 times that of water, depending on the conditions. In fact, a quick perusal of the Wikipedia article on viscosity shows that viscosities can vary by a fantastic amount — some 13 orders of magnitude, from easy-to-move gases to thick pitch that behaves like a solid except on long time scales. The situation is even more complicated than this, as some fluids can have a viscosity that changes depending on the flow. I wanted to find a way to measure the viscosities of the stuff around me, so I made the viscometer pictured below for about $1.75 (the vending machines in Clark Hall are pretty expensive).
To do this, I
Enjoyed the crisp, refreshing taste of Diet Pepsi from a 20 oz bottle (come on, sponsorships).
Cut the top and bottom off the bottle, so all that was left was a straight tube.
Mounted the bottle with on top of a small piece of flat plastic.
Mounted a single-tubed coffee stirrer horizontally out of the bottle (I placed the end towards the middle of the bottle to avoid end effects).
Epoxied or glued the entire edge shut.
Marked evenly-spaced lines on the side of the bottle.
I can load my “sample” fluid in the top of the Pepsi bottle, and time how long it takes for the sample level to drop to a certain point. A more viscous fluid will take more time to leave the bottle, with the time directly proportional to the viscosity. (This is a consequence of Stokes flow and the equation for flow in a pipe. It will always be true, as long as my fluid is viscous enough and my apparatus isn’t too big.)
So we’re done! All we need to do is calibrate our viscometer with one sample, measure the times, and then we can go out and measure stuff in the world! No need to stick around for the boring calculations! We can do some fun science over the next few blog posts!
But this is a physics blog written by a bunch of grad students, so I’m assuming that a few of you want to see the details. (I won’t judge you if you don’t though.) If we think about the problem for a bit, we basically have flow of a liquid through a pipe (i.e. the coffee stirrer), plus a bunch of other crap which hopefully doesn’t matter much.
We first need to think about how
the fluid moves. We want to find the velocity of the fluid at every
position. This is best cast in the language of vector calculus — we
have a (vector) velocity field
The first equation basically
says that we don’t have any fluid appearing or disappearing out of
nowhere, and the second is basically
I won’t drag you through the
unless you want to see them, but it turns out that for my system the
height of the fluid
[For my viscometer, the coffee stirrer has length 13.34 cm and inside diameter 2.4 mm, and the Pepsi bottle has a cross-sectional area of 36.3 square centimeters (3.4 cm inner radius). You can see how the timescale scales with these properties in the gory details section.]
Well, how well does it work? Above is a plot of the height of water in my viscometer versus time, with a best-fit value from the equations above. To get a sense of my random errors (such as how good I am at timing the flow), I measured this curve 5 separate times. If I take into account the uncertainties in my apparatus setup as systematic errors, I find a value for my viscosity as
The actual value of the viscosity of water at room temperature (T=25 C) is about